You need to put a shim/spacer in there for good contact. Sometimes the SPEs come with it, otherwise you can just cut a little bit of card stock.
RE: Screen Printed Electrode Edge Connector Part No.
RE: How to buy Rodeostat
You can purchase the Rodeostat on our website here: https://iorodeo.com/collections/cheapstat-open-source-potentiostat/products/potentiostat-shield
We ship internationally with DHL Worldwide Express.
Let us know if you have any other questions!
RE: Measurement Program
Hi, we have been working on the software updates and it should be ready in the next day or two. I will post here soon with more information but we are working hard on fixing the issues. Thank you so much for your patience ! We will keep you posted.
RE: International shipping options
Hi, we only use DHL Worldwide Express for international shipping. We have had problems using other shipping methods in some countries, but DHL has been very reliable with tracking packages and speedy delivery. I am not aware of any other international rates that DHL offer, other than Express. I understand that shipping can be too expensive.
RE: transilluminator scratches
That is a good question ! We haven't tried using glass instead of acrylic for our transilluminator. However, you could certainly put your gel onto a small glass plate over the clear acrylic when you are cutting gels to protect the acrylic. You can look for small glass plates online e.g. McMaster-Carr and other stores. We will also have a look at finding some glass plates to recommend with our different transilluminator sizes.
If you would like a replacement for your scratched clear acrylic surface, we can send you one, no problem, just let us know and we will happily help with that !
RE: Additional tests for colorimeter (pH, alkalinity, calcium hardness, etc.)
We have instructions for importing and using any of our calibration data (including the pH data) here: https://sites.google.com/iorodeo.com/colorimeter/assays?authuser=0
Try going through these steps of downloading the data, loading into the plotting program and fitting the data before exporting.
By the way, this pH calibration data will only work for measuring pH between 6-0 and 8.2.
Let me know how it goes ...
RE: Test for turbidity
@cireyes Typically turbidity is measured by the light-scatterring properties of the solution rather than absorbance. However, we have used the colorimeter for turbidity measurements and a couple of colorimeter tests (Sulfate and Potassium) are based on turbidity.
First, you will need to make a turbidity calibration curve so that you can measure your water sample. To do this, you will need to buy a turbidity standard of known NTU (Nephelometric Turbidity Units). This turbidity standard is a solution of formazin (suspended polymer particles). You can buy these online from many different suppliers such as Hach etc. I bought a 4,000 NTU standard in 2012 from Grainger, Part # 8UVJ9.
Using this NTU standard you can prepare a calibration curve in the range that you want to measure your sample. For example, in the graph below you can see that I made a range of 0 - 4000 NTU by diluting the standard with distilled water. The Red LED on the colorimeter is used to measure absorbance.
Once you have your calibration curve, you can measure your water sample to determine turbidity !
RE: Mini gel electrophoresis box
@yogeshkd The mini LED transilluminator does fit under the electrophoresis box. I have attached an image for example below. You can use this set-up to visualize gels as they are running if you place the amber lid from the transilluminator onto the gel box lid or use amber glasses. The gel box color does not matter, so you can choose either blue, green or amber. Hope that helps !
RE: Other LED colors?
We have made custom versions of the LED transilluminator with different LED wavelengths. We just made one with 400 nm LEDs for example, and we have also made one with green LEDs (523 nm). We are planning to make a white light version shortly for viewing/imaging visibly stained gels.
As far as the near infra red (700 nm) wavelengths, I don't think we have made this before but it is possible. The trick is finding the right LED. For the transilluminator boards that we use, the LED would need to be a through-hole LED with standard 3 mm or 5 mm LED pin spacing, in the wavelength of your choice and have a viewing angle of 120 degrees or more (for uniform lighting).