I am trying to get my Rodeostat to work as I would like it to. Results are however very different from what I expect them to be.
First of all, running tests seem to damage the electrode (starts bubbling + silver reference on screen printed electrode starts to become grey). I read that this might result from a high current in the beginning and I might be able to fix that using shift to 0, but I wonder why this is not happening on other potentiostats and if that really explains what I am observing (considering that there should not be any significant current on the reference electrode),
Also, the currents that I get do not make any sense when I compare them with the results I get from running a test on an Autolab PGstat204.
I would be happy to hear if someone has encountered similiar issues and can tell me what to do about it!
Thanks a lot!
Dropsens C110 electrodes